V5 pro manual
V5pro combines HIFU, RF, 40kHZ ultrasonic sounding, BIO micro current, vacuum negative pressure, Infrared LED technology and other multi-functional comprehensive instruments, can achieve body shaping, tightening and lifting the skin is a cosmetological device designed for professional work in beauty salons.
By turning the energy source into a hot condensation point, deep into the subcutaneous fat layer, destroy excess fat cells, reduce excess fat cells, stimulate collagen contraction and thickening, stimulate elastic fibers, and decompose excess fat cells into Triglyceride is excreted through the body's metabolism to reach the role of shaping.
Using radio frequency technology, the epidermis is passed directly to the subcutaneous tissue to resonate the water molecules in the skin and subcutaneous tissue, so that the tissue friction accelerates to generate heat. The increase in temperature promotes the catabolism of subcutaneous fat and stimulates collagen and elastic fibers. Proliferation and reorganization, to achieve the effect of tightening the skin and shaping and lifting;
V10 head technology
The handle incorporates a variety of black technology multi-function combination, using two 40KHZ super sound popping, two BIO micro current, two RF, vacuum negative pressure and infrared LED technology, first use 40K grease, then use high frequency RF waves Accurately acting on the water molecules in the subcutaneous fat, causing the bipolar water molecules to produce rapid vibration. The heat generated by the high-speed rotary friction heats the subcutaneous fat tissue, and exerts special physical stimulation on the skin tissue to promote the action. The skin tissue is wrinkled, and under the vacuum negative pressure, it produces subtle mechanical stimulation to the deep skin tissue, which promotes the biochemical reaction of deep adipose tissue to accelerate fat consumption, and activates lipase rapidly under specific magnetic wave and radio frequency to accelerate fat decomposition. The fat cells are decomposed into free fatty acids and glycerol, which are excreted by human metabolism, so that the volume of the fat cells is reduced, and the orange peel tissue and the fat-soluble shaping effect are improved.
1. Clean the operating parts first before use.
2. Connect the power cord and press the circular switch button (inset) at the top of the screen to turn on the instrument;
3. Click enter to enter the operation interface;
4. Click handle(A) to enter the operation interface of the large handle (inset), HIFU energy 0-10, RF energy 0-10, up and down arrows are to increase or decrease the energy (illustration), the first use of energy is recommended from the lowest 1 Start, increase according to the ability to withstand during use, because it is normal to feel hot or stinging when using it;
5. Click time to set the operation time (inset). For example, it is recommended to operate the waist and abdomen operation time for 30-45 minutes, and set the operation time according to the size of the operation part;
6. After setting the operation time and operating energy, click PAUSE (illustration). There is a round button (illustration) above the handle. The button represents start and pause. When you press the button of the handle, you will hear the sound of “drop”. To start the work, press the button of the handle to pause the output when you need to pause the work;
Handle (B) steps
1. Click enter to enter the operation interface (inset);
2. Click handle (B) to enter the small handle interface (inset), HIFU energy 0-10, RF energy 0-10, up and down arrows are to increase or decrease the energy (illustration), the first use of energy is recommended from the lowest 1 Start, increase according to the ability to withstand during use, because it is normal when there is a sense of heat;
3. Click time to set the operation time. For example, it is recommended to operate the waist and abdomen operation time for 30-45 minutes, setting the operation time according to the size of the operation part;
4. After setting the operation time and operating energy, click PAUSE (illustration). There is a round button (illustration) above the handle. The button represents start and pause. When you press the button of the handle, you will hear the sound of “drop”. To start the work, press the button of the handle to pause the output when you need to pause the work;
V5PRO system interface icon
V10 head operation steps
1. Click time to set the operation time. The operation time is recommended to be 45-60 minutes, according to the operation department.
Bit size sets the operation time;
2. Click energy setting, RF energy is 1-10, BIO micro current energy is 1-15, 40K fat burning energy is 1-10, first use energy is recommended to start from the lowest 1 first, according to the ability to withstand during use Increase, because it is normal to have a sense of heat, current, or a squeaky sound in the process of use;
3. After setting the operation time and energy level, click the icon to turn red to represent output, and blue to stand by;
4. After the operation is completed, wipe the medium in the clean operation area and rub the moisturizing product.
1. Menstrual period is not recommended for operation;
2. It is not recommended to operate during pregnancy or during the month;
3. The body is buried around metal, plastic, silicone, etc.;
4. Medical electronic devices such as hearing aids and heart pulse devices are buried in the body;
5. Severe three-high population, mental illness patients, etc.;
6. Patients with abnormal fat metabolism, including severe hyperlipidemia or severe fatty liver
7. Patients with skin allergies or inflammation, patients with infectious diseases on the skin;
8. Long-term continuous use of anticoagulant drugs; for example: non-steroid anti-hair
Inflammatory drugs or oral contraceptives, etc., long-term use of hormone drugs;
9.12 children under the age of one year and those with irrational expectations for cosmetic effects;
1. It is not recommended to wash the hot water bath immediately after 4 hours of operation;
2. It is recommended to maintain a replenishment of water up to 2000 cc per day after operation;
3. Maintain at least 10-15 minutes of exercise per day, preferably jogging or aerobic exercise.
Maintain a good state of mind ;
4. Diet should be light, do not overeating, maintain nutritional mix;
5. Pay attention to the law of work and rest, do not drink excessively, eat late at night, etc.